iTOUCH SURE SPECIFICATIONS
Max Intensity 100mA zero to peak +ve in steps of 0.5mA Constant current over 500-1500 Ohm Channels single Waveform Asymmetrical rectangular Max Pulse energy Total output limited to 25uC per pulse Power 2xAA Alkaline OR 2 x AA NiMH Battery life At least 15 hours at 50mA 300uS 50Hz, Adjustable Timer- 10, 20, 30 45, 60, 90 min Defaults to 20 min Pause button Clock key when running Output plug Fully shielded: touch-proof mini USB Weight 90 gms without batteries Dimensions 110 x 53 x 30mm iTouch Sure Programme Settings: STRES: 50Hz, 300 uS, 5/10 sec, 20 mins URGE:10Hz, 200 uS, 5/10 sec, 20 mins. MIXED: 20Hz, 250 uS, 5/10 sec, 20 mins TONE: 35Hz, 250 uS, 3/6 sec, 20 mins
Please note, due to the nature of this item we are unable to accept returns.
Do NOT use the optional electrode pads if you have a pacemaker (or have a heart rhythm problem) or with any electronic medical devices.
Do NOT use if you are or may be pregnant.
Consult your healthcare professional before use if you have epilepsy or active or suspected cancer in the area being treated.
The stainless steel in the probe’s metal plates contain some Nickel. This could cause a reaction if you have a Nickel allergy. Alternative gold probe specially made without Nickel is available to purchase from TensCare.
Caution should be used if you have a copper IUD. If discomfort occurs, discontinue treatment immediately and ask your healthcare professional for advice.
There is a small risk of stimulating the uterine wall if the IUD is not correctly positioned.